Keep OralStat device on a flat surface all times
Remove the blue instruction label covering the wells labeled “A” and “B”
Insert the handle end of the specimen collector into each of the A wells, pushing down completely, until the ring on the collector is level with the top of the well.
This punctures the foil seal (there will be a pop) in to buffer cups
On a flat, level surface gently slide the test device back and forth for at least three (3) seconds